The Cefar TENS/NMES is a 2 channel stimulator used by patients at home and therapists in clinics to provide electrical stimulation treatments in pain management (TENS) and neuro muscular stimulation (EMS/NMES).
Mains rechargeable Lithium Polymer battery: Plug your device directly to the mains plug to charge the battery. Fast and complete charge in less than 90 minutes with a battery life of 6 to 10 hours treatment depending on parameters used.
1+1 Function: The Cefar TENS stimulator features 2 independent channels that can be managed separately. Users can select either the same program on both channels, or 2 different programs that can be used simultaneously. This allows treatment of two different body parts, or two patients simultaneously, or combines NMES and TENS as one treatment.
Compliance: Keep track of patients' home treatments by monitoring how many hours of usage since the device was reset, along with total device usage.
Painscore: Record patients' pain score in the stimulator before and after treatment, with a numeric scale (1-10)
Locking function: Lock the program to be sure your patient will only use the appropriate program.
Rotatable belt clip: Turn the belt clip to have the cable outputs on the bottom or on the top depending on the body area to stimulate.
Program mode: Create up to 5 custom programs and select your own parameters.
Acupuncture mode: Select the electro acupuncture mode for clinical use only.
Outputs: Two independent and individually adjustable channels that are electrically isolated from each other.
Pulse shape: Constant asymmetric rectangular current with pulse compensation for TENS programs to eliminate any direct current component to prevent residual polarisation at skin level. Symmetric for P10 NMES Program.
Maximum pulse intensity: 99.5 mA in Constant Current mode or 99 V in Constant Voltage mode for neurostimulation. 30 mA for electroacupuncture.
Pulse intensity increments: Manual adjustment of stimulation intensity from 0 to 99 mA in minimum increments of 0.5 mA for Constant Current Mode, 0 to 99 V in 1 V increments for Constant Voltage mode for electrostimualtion (5V for first setp). 0,1 mA – 30mA steps 0,1 mA for electroacupuncture.
Pulse width: 60 to 400 μs.
Frequency: 1-120 Hz
Maximum electrical charge per pulse: 80 micro coulombs (2 × 40 μC, compensated)